The AvCount Light for Skydrol has been specifically designed to measure the particulate of Skydrol hydraulic fluid.
Skydrol is an advanced fire resistant aviation hydraulic fluid which inhibits corrosion and prevents erosion damage to servo valves. It has been approved by most airframe manufacturers including Airbus, Boeing and BAE Systems and has been used in their products for over 40 years.
Due to the highly corrosive nature of Skydrol material this special version of the AvCount is fitted with Isolast seals and a protective inner case.
With a simple test procedure and sub 4 minute test time, operator skill or expertise is kept to a minimum.
AvCount Lite can hold up to 3 test methods with custom methods created on a PC before being uploaded.
Test progress and results for up to 6 size bands are displayed as cumulative particles/ml and ISO 4406 cleanliness codes.
ISO 11171 calibration
ISO 4406 Cleanliness Codes
Portable, compact instrument
Online/offline (bottle sample)
Stand-alone or PC controlled
Programmable via PC
Battery power optional
The AvCount Lite can analyse samples with kinematic viscosities of up to 64 mm2/s. When used in conjunction with the optional Sample Delivery System, the viscosity range is extended up to 200 mm/s.
Precision and accuracy:
AvCount Lite is fitted with a precision dual piston pump. When measuring samples from a bottle, the pump draws an aliqot of sample through the measurement cell at a consistent flow rate. When connected to a pressurised line, the pump provides accurate metering and control.
Optional Protrend Software SA1810-0 can be purchased to control an AvCount Lite from a personal computer or laptop. Local custom test methods can be created, tests run and data downloaded. Protrend supports multiple AvCount Lite units, synchronising the sampling and measurement to allow trend analysis.
Calibration and verification:
AvCount Lite utilises 16 point calibration. Calibration can be carried out either using NIST traceable standard solutions or by interfacing with a primary calibrated ‘Master’ AvCount Lite. Calibration can be carried out at a user site or at our authorised regional laboratories. Seta authorised training is required to perform a calibration. Verification can be performed using AvCount Verification Material.
|PARTICLE SIZE RANGE||ISO 11171: 4 µm(c) to >100 µm(c) (calibration for larger sizes available on request)|
|NUMBER OF MEASURING CHANNELS||6 size channels displayed on instrument, up to 16 programmable via PC 4 µm(c), 6 µm(c), 10 µm(c), 14 µm(c), 21 µm(c), 25 µm(c), 30 µm(c), 38 µm(c), 70 µm(c)|
|COUNTS PER MEASUREMENT (MAX)||600,000 per ml|
|SAMPLE VISCOSITY (MAX)||64 mm2/s (using internal pump), 200 mm2/s (pressure fed @ 3 BarG) (SA1950-0 Sample Delivery System is available as an accessory)|
|SAMPLE TEMPERATURE RANGE||Ambient 0 to 70 °C|
|RESULTS||900 result memory Date/time indexed (last 20 tests can be viewed on screen) Download to PC/laptop port or ProTrend software and print via internal printer|
|SAMPLE FLOW RATE||30 ml/min ±5 ml/min|
|TOTAL SAMPLE VOLUME USED||80 ml for ASTM D7619 & IP 565, from 20 ml for other methods (includes flush cycles)|
|VOLTAGE||12 Vdc 3A with printer on, 1A with printer off Charger/mains adaptor 100 to 240 Vac, 50/60Hz input, 12 Vdc 2.5A output Optional 12 Vdc battery and charger (SA1816-0|
|SIZE (HXWXD)||250 x 145 x 350 mm|
|DEF STAN 91-091||Aviaton Fuel Specification|
|ASTM D975||Standard Specification for Diesel Fuel Oils , biodiesel, biodiesel blend, diesel, fuel oil, petroleum and petroleum products|
|IP 565||Determination of the level of cleanliness of aviation turbine fuels – portable automatic particle counter method|
|ASTM D7619||Method for counting and sizing particles in light and middle distillate fuels, and liquid bio fuels, by automatic particle counter|
|DEF STAN 91-86||Turbine Fuel, Aviation Kerosine Type: High Flash Type|
|ISO 4406||Hydraulic fluid power – Fluids – Method for coding the level of contamination by solid particles|
|IP PM FA||Determination of the level of dispersed particles in diesel fuel – Generic Automatic Particle Counter Method|